- With Calibration cert.
When closing before a beginning by easy operation, Torque Wrench can be checked daily.
LED value shown on the display can read the numerical value without an error easily.
3 kinds of measurement mode (truck mode peak mode peak hold mode) can be changed.
It corresponds to various square Drive adjustable type by 2 kinds of adapter of an accessory (change adapter hexagonal convex adapter).
Is Manufacturer’s Suggested Retail Price in Japanese market as just for reference.
- A power supply :100V-240V
- Operating temp. range : 5 ℃-40 ℃
- Storage temp. range : 20 ℃-70 ℃ of-
- Measurement mode: Truck mode/peak mode/peak hold mode
|Models||Minimum torque capacity (N m)||The upper limit torque capacity (N m)||Measurement accuracy (the %)||Min. display||dimensions (mm)D||dimensions (mm)L||dimensions (mm)S||Method for measurement||1 scale (Nm)||Units||weight (g)|
|TTC-60||2||60||± 1%+1 scale||0.00-||30.0||155||12.7||The right direction of rotation (clockwise)||0.01||Nm||2200|
|TTC-500||20||500||± 1%+1 scale||0.0-||48.0||173||25.4||The right direction of rotation (clockwise)||0.1||Nm||2600|
|TTC-1000||50||1000||± 1%+1 scale||0.0-||48.0||173||25.4||The right direction of rotation (clockwise)||0.1||Nm||2800|
- ※ A change adapter-AC adapter isn’t included in weight .
|○ A change adapter-(1 for each)|
|Convex 12.7 mm of TTC-60:concave 6.35mm x convex 12.7 mm and concave 9.5mmx|
|TTC-500 and TTC-1000:concave 9.5mmx 25.4 mm of convex and concave 12.7mmx 25.4 mm of convex and concave 19.0mmx 25.4 mm of convex|
|O six corner convex adapter (1 for each)|
|19 mm x convex 12.7 mm of TTC-60: hexagon and 27 mm x convex 12.7 mm of hexagon|
|TTC-500, 27 mm x convex 25.4 mm of TTC-1000: hexagon and 38 mm x convex 25.4 mm of hexagon|
|O AC adapter 100-240V/50-60Hz|
|O six, bolt M10 with Sumianax 45 (including washer and nut )||4|
|○ instruction manual||1|
|○ (with traceability chart ) with Calibration cert.||1|
■ Check & Repair
* Rough, it’s treated, long leaving and the use frequency, it’s a lot, there is time when accuracy is upset by a reason. calibration is needed periodically to keep accuracy .
At least, a year, once, calibration , I’ll recommend you (not free).
* Please consult with the store you buy or our business office about handling.
* By the incompleteness and the no load if failing to tear, by which a display is indication, a numerical value difference, I don’t recover, a case and Check & Repair are needed (not free).
※ The damaged situation can’t sometimes repair it.